I have reviewed the programmers reference for this device, and have made countless searches on Google for an answer, but haven’t been able to find a solution. This should bring up a page that looks something like this. IMAGe command can save a capture of the screen image into a specified file including network storage, you might want to create a local network between the PC running the test program and the scope. Thanks for all the help guys. For Ethernet, pairs 1 and 2, and 3 and 6 are used.
|Date Added:||19 March 2010|
|File Size:||60.41 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
I don’t have this particular model but I would think that SAVe: Most Active Hardware Boards: Here are the commands I used. However, it is not labviea and it is not giving an error.
Instrument Control (GPIB, Serial, VISA, IVI)
Testing Now we need to verify that we configured everything correctly and that our cable is working okay. I used the following commands in my code the way the manual says to save waveforms which can be seen in my VI: Message 1 of 5. Message 1 of 3. Be sure that the IP address labiew Subnet mask match the numbers shown above. The main problem I am still having is: Message 4 of 5.
Ben64 The best way to say thanks is to give kudos! Thank you so much for pointing that out to me.
Back to Top 4. Message 3 of 5. I’ll try to post the solution when when I get it working.
In this dialog, we want to turn off Obtain an IP address automaticallyand switch it to Use the following IP address. A cross-over cable maps the 1,2 pair on each end to 3,6 xpo4104 the other end. Hopefully it will help someone else in the same situation at some point!
Tektronix MDO MSO DPO 2000 3000 4000 Series
You should then see a dialog that looks like the image below. Most of the commands still function properly and I’ve fixed a lot of what does not using the programming manual for the series DPO and MSO Attached is a screenshot of the code I generated for capturing the screenshots, it works great. I can not open VI stillbut you epo4104 that it saves locally scope labvview, not PC?
Press OK Accept to apply the changes.
Message 5 of For Ethernet, pairs 1 and 2, and 3 and 6 are used. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as labvew type. There are only two ways to tell somebody thanks: However, neither of these solutions seem to be available on this device Acquire Screen Capture VI.
Sorry if this is in the wrong place.
Tektronix DPO Digital Phosphor Oscilloscope – Instrument Driver – National Instruments
Back to Top 3. IMAGe command can save a capture of the screen image into a specified file including network storage, you might want to create a local network between the PC running rpo4104 test program and the scope.
Can I connect three or more devices together?